I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t seem to figure that one out. I think there should be somewhere in the block diagrams of the HP B Example. I will at least try that, I tried working with two other drivers as well and somehow sampling is way harder than sweeping. Message 1 of Sampling on HP Message 10 of
|Date Added:||10 December 2004|
|File Size:||57.34 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Sampling on HP I managedto get communication between the equipment HP and LabView. Such error can occur if configuration of GPIB is not correct e. Do you have any experience with this?
Agilent | Agilent xB/C Semiconductor Parameter Analyzers VXIplug&play Instrument Driver Read Me
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples. All of the directory paths which follow assume Windows The only input commands I did were to set up teh channel definition and source setup.
Will keep you updated if it works. This readme file contains additional information concerning the driver. Did you try other commands which take a longer time to execute? Source is included so that the driver can be modified if desired. Now I can see the measured data on the monitor of the HPB. I can now input data from Labview to HP Message 5 of 6. Message 4 of 6.
The data I want to represent is IV curves. That’s where your example comes from.
I would like to know if there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be lxbview to read out the measured data directly from the HP via LabView?
Message 1 of 6. It includes a Visual Basic include file. Message 4 of I will give it a try. Here should be what you are looking for. Message 7 of I will try nevertheless to find a solution.
Agilent Technologies / Keysight Technologies 4156B
There are several advantages to using the dll form of the driver. Message 3 of Moreover I cannot see the trace of the sweep I expected a trace of 10 MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface.
Two elements of the standard do not apply to the instrument since it is not a VXI labgiew.
I have no experience with LAbView so far. Do you think somethings wrong with the following? Norbert Norbert CEO: Message 1 of Message 6 of 6.
I also have that driver that you’re using, it would be very nice if you provided me that code.